VelocityArea Flowmetering
Flow rate or discharge in an open conduit, channel or river can be calculated with the velocityarea principle.
The velocityarea principle is based on velocity measurements in a open flow like a conduit, channel or river.
Velocities and depths across the stream are measured as indicated in the figure above. A partial discharge in a section of the stream can be calculated as
q_{n} = v_{n} a_{n} (1)
where
q_{n} = flow rate or discharge in section n (m3/s, ft3/s)
v_{n} = measured velocity in section n (m/s, ft/s)
a_{n} = area of section n (m^{2}, ft^{2})
One simple way to express the section area is
a_{n} = d_{n} (l_{n+1}  l_{n1}) / 2 (2)
The total flow in the stream can be summarized to
Q = Σ_{1}^{n} v_{n} a_{n} (3)
where
Q = summarized flow rate or discharge in the conduit (m^{3}/s, ft^{3}/s)
The accuracy of estimate depends on the profile of the conduit and the number of measurements. For conduits with regular shapes like rectangular channels a limited number of measurements are required. For irregular shapes  like natural rivers or similar  higher accuracy requires more measurements both horizontal and vertical.
Example  Computing Flow Rate in a Channel
From a conduit we have three measurements:
Measured Values  Calculated Values  

n  v (m/s)  d (m)  l (m)  a (m^{2})  q (m^{3}/s) 
0  0  0  0  
1  3  1  2  2  6 
2  4  1.5  4  3  12 
3  3  0.9  6  1.8  5.4 
4  0  0  8  
Summarized  23.4 
The section areas can be calculated like
a_{1} = (1 m) ((4 m)  (0 m)) / 2
= 2 m^{2}
a_{2} = (1.5 m) ((6 m)  (2 m)) / 2
= 3 m^{2}
a_{3} = (0.9 m) ((8 m)  (4 m)) / 2
= 1.8 m^{2}
The flow rates can be calculated as
q_{1} = (3 m/s) (2 m^{2})
= 6 m^{3}/s
q_{2} = (4 m/s) (3 m^{2})
= 12 m^{3}/s
q_{3} = (3 m/s) (1.8 m^{2})
= 5.4 m^{3}/s
The total flow can be summarized as
Q = (6 m^{3}/s) + (12 m^{3}/s) + (5.4 m^{3}/s)
= 23.4 m^{3}/s
Note  there are alternative ways to calculate the section flow rates:
Simple Average Method
Using the simple average of two successive vertical depths, their mean velocity, and the distance between them can be expressed as
q_{n to n+1 } = [(v_{n} + v_{n+1}) / 2] [(d_{n} + d_{n+1} ) / 2] (l_{n+1}  l_{n}) (4)
Midsection Method
With the midsection method, the depth and mean velocity are measured for each number of verticals along the cross section. The depth at a vertical is multiplied by the width, which extends halfway to the preceding vertical and halfway to the following vertical, to develop a crosssectional area. The section flow rate can be expressed as
q_{n} = v_{n} [((l_{n}  l_{n1}) + (l_{n+1}  l_{n})) / 2] d_{n} (5)
Related Topics

Flow Measurement
Flow metering principles  Orifice, Venturi, Flow Nozzles, Pitot Tubes, Target, Variable Area, Positive Displacement, Turbine, Vortex, Electromagnetic, Ultrasonic Doppler, Ultrasonic Timeoftravel, Mass Coriolis, Mass Thermal, Weir Vnotch, Flume Parshall and Sluice Gate flow meters and more.
Related Documents

Comparing Flowmeters
A limited comparison of flowmeter principles  regarding service, rangeability, pressure loss, typical accuracy, upstream pipe diameters, viscosity and relative costs. 
Flowmeter  Accuracy
Introduction to accuracy in flow measurement devices. 
Flowmeters  Turndown Ratios
Turndown ratio (Rangeability) can be used to compare flow measurement devices like orifices, venturi meters etc. 
Fluid Flowmeters  Comparing Types
An introduction to the different types of fluid flowmeters  Orifices, Venturies, Nozzles, Rotameters, Pitot Tubes, Calorimetrics, Turbine, Vortex, Electromagnetic, Doppler, Ultrasonic, Thermal, Coriolis. 
Hydrostatic Pressure vs. Depth
Depth and hydrostatic pressure. 
Open Channel Weirs  Volume Flow Measurements
Weirs can be used to measure flow rates in open channels and rivers  common for water supply and sewage plants. 
Orifice, Nozzle and Venturi Flow Rate Meters
The orifice, nozzle and venturi flow rate meters makes the use of the Bernoulli Equation to calculate fluid flow rate using pressure difference through obstructions in the flow. 
Sluice Gate  Volume Flow Measurements
Sluice gates can be used to control and measure volume flow rates in open channels and rivers, mainly in connection to hydro power plants. 
UTube Differential Pressure Manometers
Inclined and vertical utube manometers used to measure differential pressure in flow meters like pitot tubes, orifices and nozzles. 
Velocity Pressure Head
Dynamic pressure or velocity head. 
Volume Flow  Online Unit Converter
Convert between volume flow units like gpm, liter/sec, cfm, m^{3}/h. 
Water  Volume Flow Measurement
A manual providing guidance in selecting, managing, inspecting, and maintaining water measurement devices  describes standard methods and devices commonly used to measure irrigation water.