Flowmeter  Accuracy
Introduction to accuracy in flow measurement devices.
Accuracy indicates how close a measured value is to the accepted or true value.
 "Accuracy is the degree of conformance of a measurement to standard or true value"
A measurement system consists normally of a primary element and a secondary element. Example  in a flow measurement system the orifice is the primary element and the pressure transmitter is the secondary element. Both the orifice and the pressure transmitter influence the flow measurement total accuracy.
The accuracy of flow meters can be stated as:
 percent of full span
 percent of rate
Accuracy as Percent of Full Span
Accuracy in percent of full span can be expressed as:
a_{%} = a 100% / s (1)
where
a_{%} = relative accuracy (%)
a = +/ absolute accuracy
s = full scale of the flow meter
Example  Accuracy of Flow Meter in Percent of Full Span
If the absolute accuracy is +/ 10 kg/h  the relative accuracy at full scale 1000 kg/h can be calculated as:
a_{%} = (+/ 10 kg/h) 100% / (1000 kg/h)
= +/ 1%
The relative accuracy at full scale 100 kg/h can be calculated as:
a_{%} = (+/ 10 kg/h) 100% / (100 kg/h)
= +/ 10%
Accuracy as percent of Rate
Accuracy in percent of rate can be expressed as:
a_{%} = a 100% / q (2)
where
a_{%} = relative accuracy (%)
a = +/ absolute accuracy
q = flow rate
Example  Accuracy of Flow Meter in Percent of Rate
If the relative accuracy is +/ 1% of rate  the absolute accuracy at rate 1000 kg/h can be calculated as:
a = (+/ 1%) (1000 kg/h)
= +/ 10 kg/h
The absolute accuracy at rate 100 kg/h can be calculated as:
a = (+/ 1%) (100 kg/h)
= +/ 1 kg/h
Hysteresis
Hysteresis is the maximum difference between measurement readings at the same mechanical set point when the point is set from a value above the set point and reset from a value below the set point. Hysteresis can increase over time as a result of wear and tear of the mechanical parts.
Standard Deviation Estimate
Standard Deviation Estimate is the measure of dispersion of a set of data in its distribution about the mean of the set and can be expressed as
S = [Σ(q_{a}  q_{i})^{2} / (n  1)]^{1/2} (3)
where
S = standard deviation estimate
q_{a} = mean value of measurements
q_{i} = each individual measurement value from the set
n = number of values in the set of measurements
The mean value can be calculated as
q_{a} = Σq_{i} / n (4)
Example  Estimated Standard Deviation
The flow through a flow meter is measured to 10 kg/h, 9 kg/h and 8.5 kg/h.
The mean flow can be calculated as
q_{a} = [(10 kg/h) + (9 kg/h) + (8.5 kg/h)] / 3
= 9.2 kg/h
The standard deviation can be estimated as
S = [(((9.2 kg/h)  (10 kg/h))^{2} + ((9.2 kg/h)  (9 kg/h))^{2} + ((9.2 kg/h)  (8.5 kg/h))^{2}) / (3  1)]^{1/2}
= 0.76 kg/h
Related Topics

Flow Measurement
Flow metering principles  Orifice, Venturi, Flow Nozzles, Pitot Tubes, Target, Variable Area, Positive Displacement, Turbine, Vortex, Electromagnetic, Ultrasonic Doppler, Ultrasonic Timeoftravel, Mass Coriolis, Mass Thermal, Weir Vnotch, Flume Parshall and Sluice Gate flow meters and more.
Related Documents

California Pipe Flow Metering Method
Calculate the discharge length from the open end of a partially filled horizontal pipe. 
Comparing Flowmeters
A limited comparison of flowmeter principles  regarding service, rangeability, pressure loss, typical accuracy, upstream pipe diameters, viscosity and relative costs. 
Cryogenic Fluids and Liquefied Gas Properties
Cryogenic properties as density, boiling points and heat of evaporation for fluids like hydrogen, methane, oxygen, nitrogen, fluorine and helium. 
Electromagnetic Flowmeters
An introduction to electromagnetic flowmeters and their principles. 
Flowmeters  Turndown Ratios
Turndown ratio (Rangeability) can be used to compare flow measurement devices like orifices, venturi meters etc. 
Fluid Flowmeters  Comparing Types
An introduction to the different types of fluid flowmeters  Orifices, Venturies, Nozzles, Rotameters, Pitot Tubes, Calorimetrics, Turbine, Vortex, Electromagnetic, Doppler, Ultrasonic, Thermal, Coriolis. 
Orifice, Nozzle and Venturi Flow Rate Meters
The orifice, nozzle and venturi flow rate meters makes the use of the Bernoulli Equation to calculate fluid flow rate using pressure difference through obstructions in the flow. 
Pitot Tubes
Pitot tubes can be used to measure fluid flow velocities by measuring the difference between static and dynamic pressure in the flow. 
Process Instrumentation  ISA Codes
ISA process instrumentation codes  and combinations. 
Sluice Gate  Volume Flow Measurements
Sluice gates can be used to control and measure volume flow rates in open channels and rivers, mainly in connection to hydro power plants. 
Target  Volume Flow Meters
Introduction to the target flow meters principles. 
UTube Differential Pressure Manometers
Inclined and vertical utube manometers used to measure differential pressure in flow meters like pitot tubes, orifices and nozzles. 
Ultrasonic Doppler and Time of Flight Velocity and Volume Flow Meters
An basic introduction to the ultrasonic Doppler and Time of Flight Flow Meters. 
VelocityArea Flowmetering
Flow rate or discharge in an open conduit, channel or river can be calculated with the velocityarea principle. 
Volume Flow  Online Unit Converter
Convert between volume flow units like gpm, liter/sec, cfm, m^{3}/h. 
Vortex Flowmeters
An introduction to vortex flowmeters. 
Water  Volume Flow Measurement
A manual providing guidance in selecting, managing, inspecting, and maintaining water measurement devices  describes standard methods and devices commonly used to measure irrigation water.